Exam M

Transmit/Receive
The beam angle for the M-line can be adjusted through:
 * Steer

Sequencing
The mono (m-mode only) or duplex (m-mode plus b-mode) sequence is chosen through:
 * Mode

The line segment (gate) over which the M-mode data is acquired and the image is generated is determined by:
 * M Line
 * M Line Depth
 * M Line Size

The rate at which the line will be scanned is determined by:
 * LRP

The Line Density of the B-mode image is determined through:
 * Line Density

Display
The layout of the screen (relative position of the B-mode and M-mode images) is determined by the provisional Screen Layout. This parameter does not do anything in this version of the Exam software. The zoom of the B-mode image can be changed through the Initial Zoom. The tint of the m-mode image can be changed through Chroma. The Scroll Mode determines how the m-mode image is swept: like an oscilloscope trace, or continuously running.
 * Scroll Mode
 * Initial Zoom
 * Chroma
 * Screen Layout