SonixDAQ

Introduction
The SonixDAQ is an add on to Sonix ultrasound systems that allows the user to collect raw pre-beamformed channel data in parallel. This pre-beamformed data is the raw signal coming from each individual element from the ultrasonic transducer prior to any processing and has proven to be critical in several ultrasound-based signal processing applications.

Parallel access to pre-beamformed data enables the users to collect ultrasound data in lowest possible level. This allows the users to acquire ultrasound data at a rate of up to 20,000 frames per second which is 100 times faster than the frame rate on ultrasound systems.

The specifications of the SonixDAQ that the user should be aware of before using the device for research include:
 * Supports up to 128 channels (parallel receive) each equipped with 12-bit ADC
 * 16GB of internal memory for recording the data
 * 60MHz sampling frequency
 * Supports external triggering for synchronous data acquisition
 * USB connection for programming the DAQ and transferring data back to the PC

This page is prepared to help the user in order to setup the SonixDAQ and Sonix systems for data collection.

Important Notice
The SonixDAQ is only a receive module. The SonixDAQ does not control the transmit and has no effect on the actual imaging process using Sonix ultrasound systems. Thus, it can be used in a parallel manner while a standard ultrasound imaging is in progress. If needed, SonixDAQ can also be combined with the Texo SDK to provide control over the transmit signal.

PDF Download
Download all SonixDAQ documentation as a PDF

Main Topics

 * SonixDAQ Hardware
 * SonixDAQ Software
 * SonixDAQ Installation
 * SonixDAQ Quick Test